The chromatograph is a stationary industrial device that includes a control station (industrial computer), electronic control units, a gas flow control unit (secondary sample preparation system), a sample dispensing device and an analytical unit consisting of a thermostat unit with detectors (up to two detectors) and chromatographic columns (up to eight columns).
The configuration of the complex (type and number of used detectors and chromatographic columns) depends on the nomenclature of the determined components.
To configure and diagnose the operation of the chromatograph, the LAMS software installed
on the built-in computer is used.
On the monitor screen, it is possible to display chromatograms in real time, save chromatograms, view current and archived results of measurements of the content of components, view and edit the parameters of the chromatograph.
Measured parameters: component composition
Sample type: gases, liquefied gases, liquids up to C9
Type of detector: thermal conductivity detector (DTP), flame ionization (PID), flame photometric (PFD), helium ionization (DIG)
0-1 / 10/100% all detectors
0/10/100/1000 ppm - PID, PFD, DIG
Accuracy: accident ± 2% of the scale; PID ± 1% of span
Measurement cycle time: 2-10 min typical
It is entered in the State registry of SI of the Russian Federation No. 39413-08
Pressure and flow rate: 35-2400 kPa, 20 ... 200 ml / min
Signal Outputs: 4-20 mA
Power Supply: ~ 220/240 V, 50 Hz, 1500 W
Support gases: carrier gas H2, He, N purity 99.995 (He 99.9999% for DIG), 3-4 atm, 50 ml / min., Fuel gas PID, PFD - hydrogen 99.995%, 3-4 atm, 40 ml / min., pyrolysis gas PID - air 99.995%, 3-4 atm, 400 ml / min., instrument air 3-4 atm
Installation method and conditions: wall mounting
Management: built-in PC with Windows XP or from a remote station